Copycave is a program that runs on Windows, MacOSX and Linux.
You can fast and easily create maps of caves, rotate them in 3D and convert them into therion format.
Frequently Asked Questions:
Why is there -90 in "Adj. Vertical"?
Cavers use two different ways to note declination.
Some (e.g. Sunto) define horizontal as 0 degrees. Upwards they count positive till plus 90, downwards they use negative numbers.
(This is the default in CaveMeter and the reason for the "-90".)
The others are afraid of getting confused with writing plus and minus and define horizontal as 90 degrees.
Upwards they increase the numbers till 180, downwards they decrease. So a vertical pit has 0 degrees.
(If you want that just change the -90 in "Adj. Vertical" to 0.)
How to adjust the declination?
Take a bearing over a point A to B and afterward vice versa from B to A.
Add the two declinations and divide the result by minus two, e.g:
+8.0 plus -6.4 = +1.6
+1.6 / -2 = -0.8
"Menue" > "Adj. Vertical" and type in the right field (Additional) -0.8
How precise ist the clinometer?
Results should be clearly better then +/- 1 degree.
How to calibrate the compass?
The compass ist calibrated continuously during use.
Before starting a measure you should in addition make the movement of a lying eight link in the video.
How to adjust the compass?
If you find out that the compass systematically shows e.g. 2.5 degrees too much,
then start "Menue" < "Declination-GPS" and type the correction -2.5 in the appropriate field. There you can also compute magnetic declination on the basis of GPS.
What is magnetic declination?
The Compass does not point to North but to the magnetic north pole,
which is at the time (2013) about 500 km away from geographic north pole.
Depending on your present position on earth the difference between geographic and magnetic direction to north is different.
In many parts of Canada it is more than 20 degrees.
This difference is called magnetic declination.
According to geologic standards it is changing rapidly. For example:
1983 magnetic declination near Berchtesgaden in Germany was zero degrees and could simply be ignored.
2012 it was +2.7 degrees. That means, when the compass showed 90.0 degrees, geographic direction was 92.7 degrees.
Declination at your present position is determined with menu item Declination in CaveMeter
(GPS must be activated): Wait until present position is displayed. Then press Compute.
Declination is displayed in the left upper corner.
If now you press Save, the value is stored and will be used for further calculation of direction.
How precise ist the compass?
Results should be about +/- 2 degrees. But spikes may happen.
The shorter the red bar, the more precise is the compass.
Are there any reference points even without map?
Start the Pointing Function and wait until the GPS shows values.
Keep your present position with New -> OK and move at least 200 meters away from that point.
The value at "Degrees to Point" shows the geographic direction to the kept point.
If the compass bearing differs clearly from the GPS bearing to that point then either calibrate the compass link in the video
or change the declination value with "Declination GPS".
Where are the results of the measures?
The results are saved on the SD card in the folder "cavemeter".
If you did not enter a cavename the files name is "cave.txt", otherwise it is the name you entered.
If no extension .txt was stated it will be added automatically by the program.
So it is easier to copy and paste survey data to a program for drawing the map.
How to draw a map of the cave?
The results are saved in pure text and can be used without any changes by copy&paste in programs for drawing cave maps like Therion or
GPS altitude more precise. Timedifference System-GPS with three decimal places. Illuminance in lux added at "CaveName" and light changed.
Barometer and altimeter with recording option.
Map added showing GPX points.
GPS accuracy increased to seven positions after decimal point.
Confirmation when deleting waypoints or saving existing names.
Waypoints are now saved in favourites.gpx and can be deleted from there again.
Function Direction can now load waypoints from a GPX-File (cavemeter/favourites.gpx) and thereby use e.g. data from navigation software like OsmAnd.
Menu button added.
Removed a bug.
Overview map of GPS satellites.
"Pointing" renamed to "Direction".
Drafts numbering revised.
Value for signal to noise ratio of GPS satellites is displayed.
Option to select a startscreen.
Added menue item "Help".
"Set Time" available in "Position".
Function Pointing simplified and look and feel of newer Android versions activated.
At each part of the survey any number of drafts can be created and saved.
On rooted devices time can be set from GPS signal in pointing menue.
Improved keyboard function.
A non-dazzling torch function that displays battery voltage, level and temperature.
A red bar indicates the quality of compass measuring.
Inclination calculated from GPS position is compared to the inclination from the measure.
The closer the two values are, the better the compass works.
Touch a measuring value to hold it. Border areas do not cause unintended measuring any more.
Improved energy saving.
Added browser for cave files. File extension .txt is now mandatory.
Please rename older files you want to use furthermore.
Use of umlauts in comments improved.
Help can now be achieved by pressing the magnifier button.
Functions "Pointing" and "Declination GPS" are now more clearly arranged.
Added correction value for tilting.
The direction from present position to any point in the world can now be computed, assisted by GPS. This can be very helpful to check compass accuracy. (Pointing)
Besides that you can read distance, speed, bearing, accuracy, GMT and the difference to system time.
Compass declination can now be computed on the basis of GPS. And the coordinates can be saved in the caves file. (Declination GPS)
A measurement is now calculated from more samples (600).
Touching the upper third of the screen does no longer result in a measurement. So you can easier keep the smartphone from sleep mode.
Included function to save energy.
Improved horizontal measuring in the range of -75 to -45 and +45 to +75 degrees vertical declination.
Option Compass calibration switched off as it was in contradiction to automatic continuous calibration.